Payment & Shipping Terms:
|Follow Standards:||GB/T 15171、ASTM D3078||Application Products:||Soft Packaging Bag, Hard Packaging Bottle, Medicine Bottle, Metal Tank, Bubble Wrap ,composite Packaging.|
|Vacuum Degree:||0 ~ -90 KPa||Vacuum Chamber Effective Size:||Φ270 Mm X 210 Mm (H), Φ360 Mm X 585 Mm (H), Φ460 Mm X 330 Mm (H)|
Packaging Material Sealing Testing Machine Leak Tester With Vacuum
It is suitable for the sealing test of products. Through the test, it can compare and evaluate the sealing process and sealing performance of flexible packaging pieces effectively. It is an ideal testing instrument for food, plastic flexible packaging, moist towel, pharmacy and daily chemical industries.
soft packaging bag, hard packaging bottle, medicine bottle, metal tank, bubble wrap ,composite packaging.
Meet Standards : GB/T 15171,ASTM D3078
By vacuuming the vacuum chamber, the pressure difference inside and outside of the sample immersed in water was produced, and the gas inside the sample was observed to determine the sealing performance of the sample.By vacuuming the vacuum chamber, the pressure difference inside and outside of the sample was produced, and the shape recovery of the sample after expansion and release of the vacuum was observed to determine the sealing performance of the sample.
The interior use imported SMC pneumatic components
2 pieces of filter pressure reducing valve, vacuum generator, liquid crystal pressure gauge, air filter and two-way electromagnetic valve) vacuum generator produces a vacuum, forming a stable and rapid test environment.
The basic applications:
sealing test of glass bottle, tube, tank and box
plastic bottle, pipe, metal material bottle, paper-plastic composite bag and box materials
sealing test of pen core, electronic components,
Test method :
According to GB/T 15171, to put a proper amount of distilled water in the vacuum chamber and then submerge it in the water.At this point, the distance between the top of the sample and the water surface shall not be less than 25 mm.Cover the sealed cover of the vacuum chamber and adjust to one of the following values: 20, 30, 50, 90 kPa, and keep the vacuum level at 30 s.The set vacuum value shall be determined according to the characteristics of the sample (such as packing materials used, sealing conditions, etc.) or in accordance with the relevant product standards.
During the test, attention should be paid to the leakage of the sample during the forming of vacuum and the maintenance of vacuum, and to observe whether there are continuous bubbles. Isolated bubbles are not regarded as leaks.The quality of packaging bags is measured by the measurement of sealing, and then the rationality of packaging technology is determined.
Seal Leak Tester Picture :
Contact Person: woke