Payment & Shipping Terms:
|Standard:||GB T 36794-2018; HG∕T 5052-2016||Temperature Range:||RT+5~200℃ / RT+5~300℃|
|Heating Rate:||0.5~5℃/min Adjustable Linear Heating||Temperature Display Accuracy:||0.1℃|
|Power:||1PH，AC220V，50/60Hz，10A||Temperature Control Accuracy:||±0.3 ℃|
|Distribution Accuracy:||±3℃（Within 300℃）||Port:||Shenzhen|
|Stations:||10||Display Method:||Independent Display Of Time And Temperature|
Slope Heating Retention Test Machine,
10 Working Stations Holding Power Tester
200°C / 300°C Slope Heating Retention Test Machine
|Control method||Microcomputer P.I.D. automatic temperature control calculation|
|Heating rate||0.5~5℃/min Adjustable linear heating|
|Temperature display accuracy||0.1℃|
|Temperature control accuracy||±0.3 ℃|
|Distribution accuracy||±2℃(Within 200℃),±3℃(Within 300℃)|
|Standard configuration||Weight (1kg) *10 pcs, Weight (500g) *10 pcs, test steel plate (50*75mm), 10 pcs connect plates|
|Timer function||Automatic retention time, PLC touch timing function, can record the time and temperature of sample destruction at the same time, a total of 10 groups|
|Safety protection||EGO over-temperature power-off/safe overload switch/automatically cut off heat when all specimens fall|
|Manufacturing material||Internal SUS#304 stainless steel plate, external high-grade paint treatment|
PT-6013-10 Oven Type Holding Power Tester (10 workstations) is also called adhesive tape shear failure temperature tester, slope heating retention test machine, this machine is dedicated to the determination of adhesive tape shear failure temperature. By controlling the heating rate of the oven, the time taken to break the adhesive tape specimen’s stickiness and the temperature of shear failure are measured.
1. The inner high-quality stainless steel, the outer plate is electrostatic sprayed, easy to clean.
2. PLC timing control can record the time and temperature of sample destruction at the same time, reducing the production of conventional timer consumables.
3. The temperature control system adopts adjustable linear heating, which can adapt to different test requirements.
4. Reasonable air duct design to avoid the circulating air in the box from blowing directly on the sample and affecting the test results.
Contact Person: Andia